A current sensor based on the Hall effect measures the current in the wire through the sensor opening. It does not require breaking the cable and installing terminals.
Lan Controller v3.0/3.5+(3.5, 3.6, 3.7, 3.8) only measures direct current with this sensor.
When used with Lan Controller v3.0, the sensor should be powered from 3.3V from the IDC10-1 connector and the INP1-4A inputs used.
When used with Lan Controller v3.5+ the sensor should be powered from 3.3V from the IDC10-1 connector and the INPA1-4 inputs used or powered from 5V and the INPA1-2 inputs used.
Before use, the indication should be calibrated in the Inputs tab on the Lan Controller page.
|Maximum diameter of the measuring cable||9mm|
|Power supply||3.3V or 5V (from Lan Controller)|
|Breakdown voltage (insulation)||4000V|
|Length of connecting cable||50cm|